Related Product Information
The Vybrant® DyeCycle™ Ruby stain from Invitrogen offers near-infrared emission for DNA content analysis in living cells. The Vybrant® DyeCycle™ Ruby stain is cell membranepermeant, DNA-selective, and essentially nonfluorescent until bound to double-stranded DNA. This stain takes advantage of the commonly available 488 nm and 633/5 nm excitation sources with emission >670 nm, placing cell cycle studies within reach of all flow cytometrists. Vybrant® DyeCycle™ Ruby stain may also be used with any excitation source from 488 nm through to 690 nm wavelengths. Staining protocol is simple; the suspended cells are incubated in the presence of Vybrant® DyeCycle™ Ruby stain and fluorescence is measured directly—no additional treatment or centrifugation is required. The Vybrant® DyeCycle™ Ruby stain allows simultaneous co-staining of the cell population for other parameters and offers the possibility of cell sorting based on DNA content.
The fluorescence excitation and emission spectra of the stain are shown in Figure 1 and were obtained from samples of the dye bound to DNA. The Vybrant® DyeCycle™ Ruby stain/ DNA complex has fluorescence excitation and emission maxima of 638 nm and 686 nm, respectively.
|Figure 1. Fluorescence excitation and emission spectra of Vybrant® DyeCycle™ Ruby stain bound to DNA.|
|Vybrant® DyeCycle™ Ruby stain|
2.5 mM solution
|When stored as directed, this kit is stable for at least 1 year.|
Number of assays: Sufficient material is supplied for 400 assays (Cat. no. V10273) or 100 assays (Cat. no. V10309) based on the protocol below.
Approximate fluorescence excitation/emission maxima: Vybrant® DyeCycle™ Ruby stain: 638/686 in nm, bound to DNA.
Materials Required but Not Provided
- Cells and culture medium or Hanks’ Balanced Salt Solution (HBSS)
- Flow cytometry tubes
No data are available addressing the mutagenicity or toxicity of Vybrant® DyeCycle™ Ruby stain. Since Vybrant® DyeCycle™ Ruby stain binds to nucleic acids, treat the stain as a potential mutagen and use with appropriate care. Handle the DMSO dye solution with particular caution as DMSO is known to facilitate the entry of organic molecules into tissues. Always wear suitable protective clothing, gloves, and eye/face protection when handling this reagent. Dispose of the reagent in compliance with all pertaining local regulations.
For optimal DNA content cell cycle analysis, follow these guidelines:
- Eliminate cell clumps and aggregates from the cell suspension before staining, and eliminate or correct for cell aggregates during analysis using gating or modeling software
- Staining with Vybrant® DyeCycle™ Ruby stain may be performed at room temperature with the staining time about twice as long
- Hanks’ Balanced Salt Solution (HBSS) is recommended if media is not used, do not use phosphate buffers
- Do not use glass containers with Vybrant® DyeCycle™ Ruby stain
- Do not wash or fix cells after staining with Vybrant® DyeCycle™ Ruby stain
- Eliminate dead cells from the DNA content analysis of living cells by using a dead cell discriminating stain such as SYTOX® Green, SYTOX® Blue, SYTOX® Red, or SYTOX® AADvanced™ dead cell stains, or LIVE/DEAD® Fixable Dead cell stains such as Blue, Violet, Aqua, Green, Red, or Far Red kits
- Validate flow cytometry instrument performance on the day of use
- Use linear amplification for DNA content
- Use low flow rate for acquisition
- Collect adequate numbers of events for intended application
Vybrant® DyeCycle™ Ruby Cell Staining Protocol
This staining protocol is optimized using Jurkat cells suspended in complete medium (RPMI/10% fetal bovine serum) and stained with Vybrant® DyeCycle™ Ruby stain at 37˚C.
- Remove the Vybrant® DyeCycle™ Ruby stain from the freezer and allow the stain to equilibrate to room temperature.
- Prepare flow cytometry tubes each containing 0.5 mL of cell suspension in complete media at a concentration of 5 × 105 cells/mL.
- To each tube, add 1 μL of Vybrant® DyeCycle™ Ruby stain and mix well. Final stain concentration is 5 μM.
- Incubate at 37˚C for 15–30 minutes, protected from light.
- Analyze without washing cells on a flow cytometer using 488 nm or 633/5 nm excitation and >670 nm emission (Figure 2). The Vybrant® DyeCycle™ Ruby stain may be excited with any excitation source from 488 nm through 690 nm wavelengths.
Figure 2. Histogram of live Jurkat cells stained with Vybrant® DyeCycle™ Ruby stain showing DNA content distribution. G0/G1 and G2/M phase histogram peaks are separated by the S-phase distribution. Panel A shows the distribution of this population of cells when 488 nm excitation was used with a 695/40 bandpass filter. Panel B shows the same population when 633 nm excitation was used with a 695/40 bandpass filter.
2. Practical Flow Cytometry, 4th Ed., Shapiro H. M., Ed. (2003)
3. Methods Mol Biol 281, 301 (2004)
4. Cytometry A 58, 21 (2004).